Feline Triple Test FPV FHV FCV Antibody Combo Rapid Test is a lateral flow immunochromatographic assay for the semi-quantitative detection of FPV-FHV-FCV Antibody (FPV-FHV-FCV Ab) in cat’s serum, or plasma.
Feline Triple Test FPV FHV FCV Antibody Combo Rapid test is a lateral flow immunochromatographic assay for the semi-quantitative detection of FPV-FHV-FCV Antibody (FPV-FHV-FCV Ab) in cat’s serum, or plasma.
|Product Name||FelineTriple Test FPV FHV FCV Combo Rapid Test Kit|
|Specimen||serum or plasma|
|Assay Time||5-10 minutes|
|MOQ||800 test kit|
|Delivery Time||1 week after get payment|
|Packing||10 test kits/ packing box|
|Storage Temperature||2-30°C, do not freeze, do not store the test kit in direct sunlight|
|Shelf Life||18 months|
2. Simple operation, non-professional on-site inspection.
3. The detection process is fast, and the result will be displayed in about 15 minutes.
4. The test time is less than or equal to 10 seconds, and it can detect 270 samples per hour at the fastest.
5. The detection equipment is compact, easy to carry, mobile terminal operation, cloud platform quality control.
6. The detection sensitivity is high, comparable to that of enzyme-labeled kits.
FelineTriple Test FPV FHV FCV Combo Rapid Test Kit is based on sandwich lateral flow immunolayer chromatography analysis. The test card has a test window for observing the analysis operation and reading the results. Before running the test, the test window has an invisible T (test) area and C (control) area. When the processed sample is applied to the sample hole on the device, the liquid will flow laterally across the surface of the test strip and react with the pre-coated FPV-FHV-FCV recombinant antigen. If there are FPV-FHV-FCV antibodies in the specimen, a visible T line will appear. The C line should always appear after applying the sample, which indicates a valid result. In this way, the device can show the presence of FPV-FHV-FCV antibodies in the specimen.
Feline Test Kits
Feline Panleucopenia Antigen Test
Feline Panleucopenia Virus/Coronavirus/Giardia Antigen Combo Test
Feline Infectious Peritonitis Antibody Test
Feline Herpesvirus Type-1 Antigen Rapid Test
Feline Immunodeficiency Virus Antibody/Leukemia Virus Antigen Combo Test
Feline Calicivirus/ Herpesvirus /Coronavirus/Giardia Antigen Combo Test
Feline Leukemia Virus Antigen Test
Feline Panleucopenia Antibody Test
Feline Immuno deficiency Virus Antibody Test
Feline Herpesvirus Type-1 Antibody Rapid Test
Feline Coronavirus Antigen Test
Feline Calicivirus Antibody Rapid Test
Feline Calicivirus Antigen Test
Feline Pregnancy Relaxin Test
Canine/Feline Toxoplasma IgG/IgM Antibody Test
Feline Herpesvirus/Calicivirus Antigen Combo Test
Feline Giardia Antigen Test
Feline Panleucopenia/Herpesvirus/Calicivirus Antibody Combo Test
Feline Coronavirus Anbody Test
Feline Panereatie Lipase Rapid Test
Feline Immunodeficiency Virus Antibody/Leukemia Virus Antigen /Heartworm Antigen Combo Test
Feline Immunodeficiency Virus Antibody – Feline Leukemia Virus Antigen – Feline Coronavirus Antibody –Heartworm Antigen Combo Test
Canine / Feline Combo-Test Kits
Helicobacter pylori Antigen Rapid Test
Giardia Antigen Test
Mycoplasma Pneumonia Antigen Rapid Test
Toxoplasma IgG/IgM Antibody Test
-Collect dog’s ocular, nasal or anus secretions with the cotton SWAB and make the SWAB wet sufficiently.
-Insert the SWAB into the provided assay buffer tube. Agitates it to get efficient sample extraction.
-If using serum or plasma specimen, use the dropper to collect the specimen.
-Place 3 drops of serum or plasma into the assay buffer tube for dilution. The diluted sample is used as sample extraction.
-Take out the test card from the foil pouch and place it horizontally.
-Suck the treated sample extraction from the assay buffer tube and place 3 drops into each sample hole “S” of the test card.
-Interpret the result in 5-10 minutes. Result after 10 minutes is considered as invalid.
- Positive (+): Whether the T line is clear or fuzzy, the "C" line and the region "T" line are present.
- Negative (-): Shows only the sharp C line. No T line.
- Invalid: no color lines in zone C. Whether the T line appears or not.
-All reagents must be tested at room temperature.
-Do not take the test tape out of the bag before use.
-Do not use the test after the expiration date.
-The components in this kit have been used as a standard batch processing unit for quality control testing. Do not mix parts of different batch numbers.
-All specimens are potentially infected. Must be treated strictly in accordance with the local country's rules and regulations.
Canivet cdv-cav-cvi-cpiv antigen combination test is only used for in vitro veterinary diagnosis. All results should be considered with other clinical data and can be discussed with the veterinarian. When a positive result is detected, it is recommended to use further confirmation methods such as RT-PCR.
Add: Building 2, Xinmao Qilu Science Technology Industrial Park, Tianqiao District, Jinan City, Shandong Province,China.