FPL Test feline Panereatie Lipase Rapid Test is a rapid test strip developed based on immunochromatographic colloidal gold technology for quantitatively detecting the concentration of pancreatic lipase in cat blood. This product provides a basis for the clinical diagnosis of feline pancreatitis by measuring the concentration of feline pancreatic lipase in cat blood samples. Detection limit: 3μg/L~600μg/L Detection time: 10 minutes Test samples: serum, plasma
|FPL Test feline Panereatie Lipase Rapid Test|
Serum, plasma or whole blood
800 test kits
1 week after Get payment
10 test kits/Packing box
1. One-step whole blood, serum and plasma testing.
2. Simple operation, non-professional on-site inspection.
3. The detection process is fast, and the result will be displayed in about 15 minutes.
4. The test time is less than or equal to 10 seconds, and it can detect 270 samples per hour at the fastest.
5. The detection equipment is compact, easy to carry, mobile terminal operation, cloud platform quality control.
6. The detection sensitivity is high, comparable to that of enzyme-labeled kits.
• Test devices
• Single Buffer
This product uses rapid immunochromatography technology to detect the concentration of cat pancreatic lipase. After adding the sample to the sample well of the test paper card, if there is pancreatic lipase in the sample, and the concentration of pancreatic lipase is greater than or equal to 3 μg/L, the pancreatic lipase will bind to the corresponding specific antibody labeled with colloidal gold to form a complex along the layer. The chromatographic membrane moves and is captured by the pancrelipase-specific antibody pre-coated on the chromatographic membrane, forming a wine-red detection line at the T position of the stuck shell, and the pancrelipase concentration result can be output by the ICA image analyzer. In addition, a C line is also designed in the system to verify the validity of the experiment. Regardless of negative or positive results, the line should be colored, otherwise it will be judged as an invalid result.
Strage and stablity
The kit can be stored at room temperature (2-30°C). The test kit is stable through the expiration date (18 months) marked on the package label. DO NOT FREEZE. Do not store the test kit in direct sunlight.
Feline Test Kits
Feline Panleucopenia Antigen Test
Feline Panleucopenia Virus/Coronavirus/Giardia Antigen Combo Test
Feline Infectious Peritonitis Antibody Test
Feline Herpesvirus Type-1 Antigen Rapid Test
Feline Immunodeficiency Virus Antibody/Leukemia Virus Antigen Combo Test
Feline Calicivirus/ Herpesvirus /Coronavirus/Giardia Antigen Combo Test
Feline Leukemia Virus Antigen Test
Feline Panleucopenia Antibody Test
Feline Immuno deficiency Virus Antibody Test
Feline Herpesvirus Type-1 Antibody Rapid Test
Feline Coronavirus Antigen Test
Feline Calicivirus Antibody Rapid Test
Feline Calicivirus Antigen Test
Feline Pregnancy Relaxin Test
Canine/Feline Toxoplasma IgG/IgM Antibody Test
Feline Herpesvirus/Calicivirus Antigen Combo Test
Feline Giardia Antigen Test
Feline Panleucopenia/Herpesvirus/Calicivirus Antibody Combo Test
Feline Coronavirus Anbody Test
Feline Panereatie Lipase Rapid Test
Feline Immunodeficiency Virus Antibody/Leukemia Virus Antigen /Heartworm Antigen Combo Test
Feline Immunodeficiency Virus Antibody – Feline Leukemia Virus Antigen – Feline Coronavirus Antibody –Heartworm Antigen Combo Test
Canine / Feline Test Kits
Helicobacter pylori Antigen Rapid Test
Giardia Antigen Test
Mycoplasma Pneumonia Antigen Rapid Test
Toxoplasma IgG/IgM Antibody Test
Collect cattle’s fresh whole blood, and get the serum or plasma specimen for use. Immediately apply the testing once collected the specimen.
Take out the test device from the foil pouch and place it horizontally.
Place 2-3 drop (around 70-100uL) of serum or plasma specimen into the sample hole. Start timer and wait for result.
Interpret the result in 5-10 minutes. Result after 15 minutes is considered as invalid.
INTERPRETATION OF ASSAY RESULT
- Positive (+): Whether the T line is clear or fuzzy, the "C" line and the region "T" line are present.
- Negative (-): Shows only the sharp C line. No T line.
- Invalid: no color lines in zone C. Whether the T line appears or not.
Add: Building 2, Xinmao Qilu Science Technology Industrial Park, Tianqiao District, Jinan City, Shandong Province,China.